Send in your spam and get the offenders listed
Create a rule in outlook or simply forward the spam you receive to email@example.com
AUDIENCE: Risk Manager, Nursing, Anesthesiology, Otolaryngology, Surgery, Critical Care Medicine
ISSUE: Medtronic announced that on May 8, 2015, it began notifying hospitals and distributors worldwide that affected lots of its Covidien Shiley tracheostomy tubes were formed with a wider-angle bend than standard models manufactured after November 29, 2012. The company initiated the field action following customer complaints that included reports of 12 serious patient injuries, such as breathing difficulties that impacted oxygen levels, immediately upon tube placement or discomfort. Replacement of the tracheostomy tube with product manufactured prior to November 29, 2012 addressed the patient breathing difficulty or discomfort.
The recall includes specific lots from eight product lines that were manufactured after November 29, 2012. See the Press Release for detailed product recall information.
BACKGROUND: A Shiley tracheostomy tube is put through a patient's trachea (windpipe) during a tracheostomy procedure to help provide an airway and facilitate the ability to breathe.
RECOMMENDATION: Customers and distributors who have provided the recalled Shiley tracheostomy tubes to a homecare provider or patient must notify the primary care physician and the homecare provider that these products should be discontinued from use and returned. If one of the recalled products is currently in use in a homecare patient, and the patient is not experiencing any discomfort, breathing difficulties or any other issues related to the tube, it is recommended that the patient's physician evaluate the continued use. If the physician advises leaving the tracheostomy tube in place, the tube must be replaced with an alternate device at the next tube exchange.
Healthcare professionals and patients are encouraged to report adverse events or side effects related to the use of these products to the FDA's MedWatch Safety Information and Adverse Event Reporting Program:
Read the MedWatch safety alert, including a link to the press release, at:
All titles, content, publisher names, trademarks, artwork, and associated imagery are trademarks and/or copyright material of their respective owners. All rights reserved. The Spam Archive website contains material for general information purposes only. It has been written for the purpose of providing information and historical reference containing in the main instances of business or commercial spam.
Many of the messages in Spamdex's archive contain forged headers in one form or another. The fact that an email claims to have come from one email address or another does not mean it actually originated at that address! Please use spamdex responsibly.
Google + Spam © 2010- 2017 Spamdex - The Spam Archive for the internet. unsolicited electric messages (spam) archived for posterity. Link to us and help promote Spamdex as a means of forcing Spammers to re-think the amount of spam they send us.
Our inspiration is the "Internet Archive" USA. "Libraries exist to preserve society's cultural artefacts and to provide access to them. If libraries are to continue to foster education and scholarship in this era of digital technology, it's essential for them to extend those functions into the digital world." This is our library of unsolicited emails from around the world. See https://archive.org. Spamdex is in no way associated though. Supporters and members of http://spam.abuse.net Helping rid the internet of spam, one email at a time. Working with Inernet Aware to improve user knowlegde on keeping safe online. Many thanks to all our supporters including Vanilla Circus for providing SEO advice and other content syndication help | Link to us | Terms | Privacy | Cookies | Complaints | Copyright | Spam emails / ICO | Spam images | Sitemap | All hosting and cloud migration by Cloudworks.