Send in your spam and get the offenders listed
Create a rule in outlook or simply forward the spam you receive to firstname.lastname@example.org
ISSUE: FDA is providing notification of a Class I recall of FLOW-i Anesthesia Systems by Maquet. The firm has received several complaints where patient cassettes, which are the center of gas flow in the system, have come loose. The patient cassette locking device may accidentally release the patient cassette from its mount when users perform a change of system. This may cause anesthesia gas to leak and could prevent the ventilator from providing breathing support if not corrected immediately.
The firm has received 10 reports this device has malfunctioned; no injuries or deathshave been reported.
BACKGROUND: The Flow-i Anesthesia System administers anesthesia while providing ventilation to patients with no or limited ability to breathe. The system is used in hospitals, for use in a range of patients from neonatal to adult.
RECOMMENDATION: A Maquet Service representative will contact consignees to arrange replacement of the FLOW-i Anesthesia System patient cassette locking device.
For questions regarding this field action, please contact a Maquet Service Representative or Maquet Technical Support at 1-888-627-8383 (Press option 3, followed by option 1 and then option 1 again), Monday through Friday, between the hours of 8:00 a.m. and 5:00 p.m. EST.
Healthcare professionals and patients are encouraged to report adverse events or side effects related to the use of these products to the FDA's MedWatch Safety Information and Adverse Event Reporting Program:
Read the MedWatch safety alert, including links to the Class I Recall Notice, at http://www.fda.gov/Safety/MedWatch/SafetyInformation/SafetyAlertsforHumanMedicalProducts/ucm453524.htm
You are subscribed to MedWatch Safety Alerts for U.S. Food & Drug Administration (FDA). This information has recently been updated, and is now available.
All titles, content, publisher names, trademarks, artwork, and associated imagery are trademarks and/or copyright material of their respective owners. All rights reserved. The Spam Archive website contains material for general information purposes only. It has been written for the purpose of providing information and historical reference containing in the main instances of business or commercial spam.
Many of the messages in Spamdex's archive contain forged headers in one form or another. The fact that an email claims to have come from one email address or another does not mean it actually originated at that address! Please use spamdex responsibly.
Google + Spam © 2010- 2017 Spamdex - The Spam Archive for the internet. unsolicited electric messages (spam) archived for posterity. Link to us and help promote Spamdex as a means of forcing Spammers to re-think the amount of spam they send us.
Our inspiration is the "Internet Archive" USA. "Libraries exist to preserve society's cultural artefacts and to provide access to them. If libraries are to continue to foster education and scholarship in this era of digital technology, it's essential for them to extend those functions into the digital world." This is our library of unsolicited emails from around the world. See https://archive.org. Spamdex is in no way associated though. Supporters and members of http://spam.abuse.net Helping rid the internet of spam, one email at a time. Working with Inernet Aware to improve user knowlegde on keeping safe online. Many thanks to all our supporters including Vanilla Circus for providing SEO advice and other content syndication help | Link to us | Terms | Privacy | Cookies | Complaints | Copyright | Spam emails / ICO | Spam images | Sitemap | All hosting and cloud migration by Cloudworks.